Warning: Use of undefined constant medium - assumed 'medium' (this will throw an Error in a future version of PHP) in /home/uzloorg/uzlo.ru/docs/wp-content/themes/yoo_nano3_wp/warp/systems/wordpress/layouts/page.php on line 16
The unit operation principle is based on measuring the pressure drop created by the compressor, associated with the passage of air through the sample.
The process is fully automated and is controlled by only by two buttons. Measurement accuracy and ease of use of the unit are achieved due to the modern electronics and pneumatic circuit with automatic measurement mode selection.
High-precision electronic sensors in combination with pressure-feeding equipment made it possible to stop using massive bells and measuring scales. After an essential redesign of the unit, its overall dimensions were reduced by several times.Brochure
|Measurement range No.1, gas penetrability units||30-300|
|Absolute error, from 30 to 300||±5|
|Measurement range No.2, gas penetrability units||300-1000|
|Absolute error, from 300 to 1000||±20|
|Overall dimensions, mm||235x190x175|
|Weight, max. kg||5,5|
|Power voltage, V||220|
|Power voltage frequency, Hz||50|
|Power consumption, W||20|